|When you align the wind sensor, Vaisala recommends using compass binoculars (such as Minox B 7×50DCM).|
|Do not remove the instrument or sensor from the mounting kit during alignment.|
- Remove the transportation damper that protects the array and store it for future use.
- To align the sensor, use a compass or other similar method and rotate the sensor so that the arrow points North.
- Tighten the bolts of the mounting kit. Tightening torque 5 Nm.
- If you cannot align WMT700, measure the deviation angle from North and configure the wind direction offset error.