|If you have a fully-heated WTM700, use a hybrid cable.|
Vaisala provides the following open-lead hybrid cables for connections to other host systems:
|Cable 2 m||CBL210706-2MSP||Cable carries through all signals from WMT700.|
|Cable 10 m||CBL210706-10MSP|
|Cable 15 m||CBL210706-15MSP|
|Cable 26 m||CBL210706-26MSP|
|Cable with custom length||CBL210706-SPEC||Cable carries through all signals from WMT700. Define the length when you order WMT700.|
|Long cables may decrease WMT700 capabilities, see Power and cable recommendations.|
WMT700 cables are shielded hybrid cables with several different wires. The cable contains 2 x 2 x 0.14 mm2 + 7 x 0.14 mm2 + 2 x 0.25 mm2 + 4 x 1.0 mm2 wires with screen.
The sensor and the cabling must be protected with a fuse or a circuit breaker. The maximum sizes of the fuses or circuit breakers protecting the power lines:
The circuit breaker can be either 2-pole (on both positive and negative/ground) or 1-pole (positive wires). You can also replace the circuit breaker with a cartridge fuse with T (slow acting) tripping characteristics.