- Uninstall WS425.
- Run WMT700 cable through the WS425 mounting kit and connect it to WMT700.
Tighten the connector by rotating the connector by hand clockwise.
Make sure that the connector is properly tightened before proceeding to the next step. If you cannot tighten the connector, loosen the mounting adapter fixing screws, remove the mounting adapter, and connect the cable. Reattach the mounting adapter.
- Attach the WMT700 sensor body to the WS425 mounting kit and tighten the bolt.
- Remove the transportation damper protecting the array and store it for future use.
Connect the cable to the data acquisition system and power supply.
Connect the wires.
WMT700 is now ready for operation.