This procedure restores the humidity measurement accuracy of the probe. No adjustment after the sensor change is needed.
To perform this procedure, you need a new INTERCAP® sensor. It is also recommended that you replace the filter with a new one.
- Remove the filter from the probe by turning it counter-clockwise.
There are 2 sensors under the filter, the INTERCAP® sensor and a temperature sensor. Identify the INTERCAP® sensor. Do not
touch the temperature sensor.
Pull out the old INTERCAP® sensor and insert a new one. Handle the new sensor by the plastic frame.
CAUTION Do not touch the sensor plate.
- Attach a new filter on the probe.