Temperature adjustment in 1 or 2 points can be done if there is reason to believe that the adjustment is changed. If the adjustment is done in 1 point only, make sure the reference condition represent the measurement environment well.
- Detach the screw from the probe handle to expose the calibration button.
- Press the calibration button in the probe handle with a thin sharp-pointed tool, such as a small screwdriver. Pressing the button switches the indicator to adjustment mode.
- Choose T and press Select.
- The adjustment mode is now on. Press Adjust to select the adjustment method (1-point or 2-point adjustment).