To perform a 1-point adjustment of carbon dioxide (CO2), measurement using I2C protocol, you must perform the following sequence:
- Using the Adjust command, start 1-point adjustment for the chosen parameter.
- Place the sensor in the reference environment and wait until the measurement has stabilized.
- Using the Adjust command, record the measured point and give the reference value.
- Use the Adjust command to end the adjustment.