|USB connection to Insight PC software cannot be established
||USB connection has been disabled with the DIP switch on the transmitter component
||See the instructions in Disabling and enabling USB functionality and RS-485 termination with DIP switch.
|Functionalities are not available in Insight PC software.
||Insight PC software is in Basic
Mode. Certain functionalities are only available in Advanced Mode.
||Change the user mode from Basic
Mode to Advanced Mode in the Insight PC software main view Settings menu.
|Measurement performance affected after calibration.
||Probe has not been switched from calibration mode to normal operating mode in the
Insight PC software.
||Open the calibration menu in the Insight PC software (Close selection, and ensure that you confirm closing the calibration mode when exiting
with the Yes selection.
). Exit the calibration menu with the |