All Sanitary OEM Refractometers are shipped with the factory default IP address of 169.254.23.33. This address belongs to the Zeroconf addresses (as defined in IETF standard RFC 3927) so that it can easily be reached from a stand-alone computer, usually without changing the network settings of the computer.
This address remains in the instrument even after a different IP address has been set. The sensor answers in the address that is first called up after startup.
|If there is more than one Sanitary OEM Refractometer in the same network, this address cannot be used. For more information, see Configuring refractometer network.|
Change the IP address of the instrument through the instrument homepage. For more information, see Instrument homepage.