|CAUTION Only factory installed AC (mains) cables have been safety type tested and assembly tested with the product. Modifying or replacing the factory installed AC (mains) cable will void UL/SGS listing.|
|Number in figure||Wire||Min. … max. wire cross-section|
|1||Line wire||0.5 … 2.5 mm2 (20 … 14 AWG)|
- Strip 55 mm (2.17 in) of the cable to expose the wires.
Cut off 10 mm (0.39 in) of the line and neutral wires. Leave the grounding wire 55 mm
(2.17 in) long.
CAUTION Make sure that the grounding wire is longer than the line and neutral wires. Under mechanical stress, the grounding wire must be the last to disconnect from the protective ground terminal.
- Strip the ends of the individual wires to expose the conductors for a length of 8 mm (0.32 in).