The enclosure can be mounted on a wall, on a bulkhead, or to a mast using the optional mounting kit. The wiring inside the enclosure is done at the factory. AC (mains) and WMT700 wiring are completed during the installation.
In addition to items in the delivery, you need the following:
- AC (mains) cable with a minimum cross-section of 3 × 1.5 mm2 (15 AWG)
- An external disconnection device (for example, a circuit breaker) for the AC (mains) power connection
- APPK-SET mounting kit for mast installation (optional)
- Signal cable with a minimum of 3 wires and a cross‑section of 0.25 mm2 ... 1.5 mm2 / AWG 24 ... AWG 16 (for connecting PJB to weather station, optional)
The cable between the wind sensor and PJB480 is included in wind sensor delivery package.
|CAUTION Use an AC (mains) cable with a minimum cross‑section of 3 × 1.5 mm2 (15 AWG). Thinner cables can damage sensors or the system.|
|To prevent corrosion and oxidation, use copper paste or equivalent on screws and connector threads.|