|CAUTION Do not remove the filter.|
|CAUTION Do not attempt to clean the sensors under the filter in any way.|
|Avoid exposing the probe to cleaning agents for unnecessarily long periods of time.|
You can clean the H2O2 and humidity probe body and temperature probe body and cable by wiping them with a soft, lint-free cloth moistened with water or mild cleaning agent, such as isopropanol. Do not wipe the filter: wiping the filter may block its pores and/or deposit residue on the filter. If the filter is heavily contaminated, replace it.
- Avoid touching the filter. If you need to touch the filter, always wear clean gloves (cotton, rubber, or similar material). Keep the filter free of any grease or oil.
- Do not scrape the probe body.
- Do not immerse the probe in liquid to clean it.
- Wipe cleaning agents off the probe after cleaning.
If needed, you can spray the probe surfaces with water.
After cleaning the probe, it is recommended to perform a sensor purge.