The transmitter has 3 analog outputs that can be configured with the Insight PC software. Each output has the same set of configuration options in the analog output 1…3 menus under the Configure device menu (Analog output 1 shown in example).
|The available selections depend on the probe that is connected to the transmitter (for example, CO2 output parameter selection is only available for CO2 probes).
The Analog output type selection in the General menu defines whether the outputs are used for voltage or current output. When you change the output type, Insight reloads the transmitter.
Receiving analog output settings automatically from probe
- The transmitter is a new unit and has not been configured yet.
- The user has removed the analog output configuration from the transmitter.
|It is recommended to always review the analog output settings with the Insight PC software after connecting a probe.