The probe sets the analog output channel into a defined error level instead of the measured result in two situations:
- Probe detects a measurement malfunction. This means an actual measurement problem, such as sensor damage or unsuitable environmental conditions.
- Measured value(s) are significantly outside the scaled output range.
The default error level depends on the output type:
|Output||Default Error Level|
|0 ... 20 mA||23 mA|
|4 ... 20 mA||2 mA|
|0 ... 5 V||0 V|
|0 ... 10 V||0 V|
The probe resumes normal operation of the analog output when the cause of the error state is removed.