When necessary, various environmental compensations can be applied to improve the CO2 measurement accuracy of the probe.
- Background gas oxygen (O2) content
- Background gas relative humidity (%RH)
|To apply an accurate relative humidity compensation, make sure that also the temperature compensation and pressure compensation configurations match the measurement environment.|
The probe has an on-board temperature sensor that can be used to compensate for temperature. Additionally, if the probe is integrated in a system that measures one or more of the compensation parameters (T, P, RH, O2), they can be updated to the probe continuously.
Compensation parameters are configured on the order form when ordering the probe, and can later be updated using Vaisala Industrial Protocol or Modbus protocol.
You can also turn off any of the compensations. In that case, the probe uses the default compensation value that is mathematically neutral for the probe’s internal compensation model.
- Vaisala Industrial Protocol serial commands
- Vaisala Insight software
- Indigo transmitter's wireless configuration interface
- Modbus configuration registers
- MI70 hand-held indicator
|The environmental compensation configuration options available in the MI70 handheld indicator are not compatible with the probe. Using MI70 to configure the environmental compensation settings of the probe can decrease measurement accuracy.|