You can see the compensation values that
are currently used by the probe by selecting them as display quantities from Main menu > Display > Quantities and Units. The quantities are as follows:
active temperature compensation value.
active pressure compensation value.
active oxygen concentration compensation value.
active relative humidity compensation value.
You can change the compensation
settings from Main menu > Settings > Measurement settings.
When you turn a
compensation off, the probe still shows a value for the corresponding display quantity (for
example, Pcomp shows 1013.2 hPa). This is the default compensation value that is
mathematically neutral for the probe’s internal compensation model.
GMP251 cannot receive the compensation values from a second
probe connected to MI70. Only direct configuration is supported.