Use the ALARM command to change the dew point temperature above which the LED indicator plug is lit. The currently measured measurement parameter is used. You can also disable the LED indicator.
ALARM [on/off] [limit]<cr>
- Enable (on) or disable (off) the LED indicator function.
- Dew point temperature above which the LED indicator is lit. Range -20 ... 50 °C.
Example (show the current status of the LED indication):
>alarm Alarm Led : OFF Alarm level : 20.000000 >
Example (enable the LED indicator at dew point above 20 °C):
>alarm on 20 Alarm level : 20.000000 Saving settings...done >