Use the L command to view the current user adjustment parameters that are affected by the CRH, RHCAL, CT, and LI commands. This command is useful for checking the currently applied correction.
The output values are as follows:
- Cp offset and gain: capacitance, adjusted using the CRH command.
- RH offset and gain: adjusted using the RHCAL command.
- T offset and gain: adjusted using the CT command.
Example (shows default values, no user adjustment done):
>l Cp offset : 0 Cp gain : 1 RH offset : 0 RH gain : 1 T offset : 0 T gain : 1 >