- Power up the module and check the output reading, preferably using a humidity reference.
(Optional:) Connect to the service port and use the ATEST command to set the analog output to desired voltage or current. Verify
that your measurement system shows the correct reading. See Test Analog Output.
Note that ATEST is one of the advanced serial commands that must be activated using the PASS command.